Digital Camera Noise Measurements
This webpage describes a newly revised international standard, ISO 15739:2013, Photography — Electronic still picture imaging — Noise measurements. (Click to go to the US ANSI website to purchase ISO 15739.) ISO 15739:2013 was developed by ISO/TC 42. This webpage includes background information on the standard, and a link to download the software which can be used to perform the measurements defined in ISO 15739:2013.
- Why measure noise?
- Camera noise measurements
- Measurement software
- Related IS&T papers
1. Why measure noise? Noise is defined as unwanted variation in the response of an imaging system. Noise is an important attribute of digital cameras, such as digital still cameras and camera phones. ISO 15739:2013 provides users and manufacturers with a standard method for measuring this important image quality attribute.
Noise is measured using a test chart defined in ISO 14524. (Click to go to the US ANSI website to purchase ISO 14524.) Note that the image below is a representation showing one layout for an ISO 14524 test chart. The image shown is uncalibrated and cannot be used for ISO 14524 testing purposes.
OECF test chart defined in ISO 14524
2. Camera noise measurements: The visibility of noise to human observers depends on the magnitude of the noise, the apparent tone of the area containing the noise, and the spatial frequency of the noise. The magnitude of the noise present in an output representation depends on the noise present in the acquired image data and the contrast amplification or gain applied to the data in producing the output. The noise visibility is different for the luminance (or monochrome) channel and the color (or color difference) channels. ISO 15739:2013 accounts for these factors in measuring and reporting the camera noise.
Camera dynamic range measurements: Measurement of camera dynamic range is closely associated with measurement of camera noise. ISO 15739:2013 defines Digital Still Camera (DSC) dynamic range to be the ratio of the maximum exposure level that provides a pixel value below the highlight clipping value to the minimum exposure level that can be captured with an incremental signal-to-temporal-noise ratio of at least 1. ISO 15739:2013 specifies how to measure and report DSC dynamic range.
The standard includes a number of important annexes:
- Annex A specifies the method for determining the components of the digital camera noise from a number of samples.
- Annex B describes a procedure using a human visual response model as a method for weighting the spectral components of the noise.
- Annex C gives a step-by-step procedure for determining the signal to noise ratio and incremental gain.
- Annex D lists recommendations for practical viewing conditions for various output media.
Click to download a zip file of camera noise measurement software.
3. Measurement software: Experts from ISO/TC 42 have developed software for performing the noise measurements specified in Annex B of ISO 15739:2013.
Note: The software is updated as of March 2016. The correction to the software precedes the correction to ISO 15739.
4. Related papers published by IS&T: The following are links to IS&T and SPIE papers which describe the development of aspects of this noise measurement standard.
Kelly, S. C. and Keelan, B.W., ISO 12232 Revision: Determination of Chrominance Noise Weights for Noise-Based ISO Calculation, SPIE Image Quality and System Performance II, January 2005.
- Aoyama, K ., Enomoto, H. and Hung, P.C., An evaluation of scanner noise based upon a human visual model, IS&T’s 49th Annual Conference, (1996), pp. 322–324.
Holm, J. and Süsstrunk S., An EIQ-subjective image quality correlation study, IS&T’s 47th Annual Conference/ICPS (1994), vol. II, pp. 634–640.
Juenger A.K. and Higgins E.W., A graininess metric for electronic still photography, IS&T’s 46th Annual Conference, (1993), pp. 408–410.